can you plz explain the working and theory of moving coil galvanometer
A current carrying conductor placed in a magnetic field experiences a torque.
It consists of N turns of coil PQRS wound over a non-magnetic metallic frame usually rectangular or circular. The coil is suspended from a torsion head H by means of phosphor bronze strip in a magnetic pole by pieces N and S. A mirror M is attached to the phosphor bronze strip. The other end of the coil is connected to a hairspring, which in turn is connected to an external terminal.The concave or cylindrical pole pieces ensure that the plane of coil is parallel to B thus (cosq = 1 and sina = 1) in the expression t = NIBA cosq and t = NIBA sina .
The whole arrangement is enclosed in non-magnetic case, which is provided with leveling screws. The torsion head is connected to terminal T1. The galvanometer can be connected to the circuit through terminals T1 and T2.
When a current is passed through the coil in the direction PQRS, then the coil experiences a torque
t = NIBA sina
Since the magnified field is radial the plane of the coil is parallel to the magnetic field such that a = 90o and hence t = NIBA.
The coil rotates and the phosphor bronze strip gets twisted. As a result a restoring torque comes into play trying to restore the coil back to original position.
If f be the twist produced in the strip and C be the restoring torque per unit twist then the restoring torque = Cf.
where G is the galvanometer constant.
This shows that galvanometer has a linear scale and detects the presence of current.
Hope this helps
You have rated this answer /10
- Biology Question Answers for CBSE Class 12 Science
- Chemistry Question Answers for CBSE Class 12 Science
- Economics Question Answers for CBSE Class 12 Science
- Hindi Question Answers for CBSE Class 12 Science
- Mathematics Question Answers for CBSE Class 12 Science
- Physics Question Answers for CBSE Class 12 Science